SH-2 Hot Plate Magnetic Stirrer Mixer Dual Control Ocean Green & Silver-white. HIGH QUALITY - 12x12 cm Enclosed heating plate, Cast aluminium, Surface spraying, with characteristics of flame protection, fast warm-up and durability.
POWERFUL - Maximum stirring capacity can reach to 1000 ml, Maximum heating plate temperature can reach to 380? Stirring speed 1002000 rpm adjustable. EASY USE - Dual controls Heating and stirring at the same time, 1 inch magnetic stir bar is enough to stir a lot of liquid, ideal for stirring in scientific research, industry, agriculture, health and medicine, home brew, etc. RELIABLE - This lab hotplate stirrer has obtained CE safety certification. Comes with complete accessories : Adjustable Thermometer Rack, Stainless Steel Rods, Fuse, US Plug, Power Cable, Operational Manual and a Teflon-coated Stirring Bar (9 mm Diameter 28 mm length).
NOVEL DESIGN - The silver-white panel matches the Green curved lines like a gentle moonshine that falls in love with a rough sea. Tell us to be passionate and focused at work, and have fun. Voltage: 100120 watts, 60 Hertz 2. Heating power: 180 watts 3.
Stirring power: 10 watts 4. Maximum heating plate temperature: 380°C 5. Maximum Stirring capacity: 1000ml liquid (H2O) volume 7.
Round Heating plate dimensions: 120mm x 120mm (4.7 inch) 8. Heating plate material: Aluminium 9. Overall dimensions (LxWxH): 200mm x 125mm x 90mm 10. Height with support stand: about 320mm 11.
Weight with packaging: 2.2 kg? Packing list 1 x Main unit 1 x Stirring bar (Diameter 9mm x 28mm length) 1 x Power Cable with US Plug 1 x Rack with Rods 1 x Fuse 1 x User Manual? If you find the stirrer vibrates violently of stops stirring, please cut off the power and inspect the bottom of the beaker to see if it is level and if its position is proper while checking the voltage to see if it is 110V±10V. Other while the irregular phenomena like above will certainly occur.
The apparatus can work continuously for 8 hours at a medium speed and for 4 hours at a high speed. If the power is on and he indicator is not lit, please inspect whether motor when power changes and the motor is overloaded. If the fuse is damaged, replace it with the fuse supplied together with the machine or a proper one of the same specification you can find.
If the electric motor refuses to turn under the proper power after a long time of operation, please inspect whether the screws connected with the motor's axle are loose. When the magnet drops, please stick it with 502 glue available on the market. Pay attention to the cleaning and poles of the magnet. This device cannot display temperature, If you need accurate temperature, you'd better prepare a thermometer. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Lab Mixers, Shakers & Stirrers\Magnetic Stirrers & Hotplate Stirrers".
The seller is "gilmine09" and is located in this country: US. This item can be shipped to United States.